LIYI Stainless Steel Lab Sand And Dust Chamber 1000L IP56X Ingress Protection
1000L IP56X Ingress Protection Test Equipment Sand Testing Lab Sand Test Chamber is according to IEC60529, IEC 60598, GB2423.37-89 (Test L:dust test methods), GB 4208-93 protective casing grade (IP code), GB/T 4942.2-93(protective casing grade of low-voltage apparatus), GB 7000.1-1996, GB 7001-1986(protective casing grade of lamps),DIN40050 and IP5K0. The XB-OTS-IP65X is applied in LED or other luminaries for IP5X and IP6X test.
Centrifugal fan is the main source of dust movement, the high-speed operation of the fan so that dust into the fan due to the centrifugal force of the role of fan blades into the dust pipe dust. The wind speed of the pipeline is usually more than 7.5m / s (too low wind speed will make the dust precipitate in the pipeline will cause the test tube can not work). The dust line was sprayed from above into the test chamber and passed through the sample at a rate of> 2 m / s. Sand and dust chamber As the lower part of the working box for the funnel, it can precipitate down the dust into the lower part of the 72 × 50um filter. Through the vibrating motor vibration, less than the diameter of the filter hole in the dust and centrifugal fan sucked so back and forth.
1.Providing a simulated environment of dry sand blowing and dust emission.
2.Long time and continuous running without fault.
3.Ensure a certain amount of dust blowing, dust blowing speed, dust blowing time and automatically set.
4.With dust dehumidifying and controlling the temperature and humidity of the inner environment.
5.Ensure the whole test is recyclable.
6.perfect safety protection function
1. IEC 60529:2001 / GB 4208-2008 of protection provided by enclosure (IP code)
2. IEC 60034-5: 2000 / GB/T 4942.1-2006 Rotating electrical machines-Part 5: Degree of protection provided by the integral design of rotating electrical machines (IP code)-Classification, IDT
3. IEC 60068-2-68: 1994 / GB/T 2423.37-2006 Environmental testing -Part 2: Tests-Test L: Dust and sand IDT
4. IEC 60068-1:1988 / GB/T 2421-2008 Environmental testing- Part 1: General and guidance, IDT
5. IEC 60068-5-2:1990 / GB/T 2422-2012 Environmental testing-Part 5: Guide to drafting of test methods-Terms and definitions, IDT
|Inner chamber size||
|Temperature range||RT+10~70℃ (can customize, confirm before order)|
|Related humidity||45%~75%(can not display)|
|Standard wire diameter of metal mesh||50um|
|Standard spacing between lines of mesh||75um|
|Amount of talcum powder||2～4kg/m|
|Powder type||Dry talcum powder|
|Dust reduction mode||Free falling dust|
|Test time||0-999H (adjustable)|
|Machine power source||220V|
|Sample power source||220V|
|Safty function||Leakage protection, short circuit protection|
Electronic and Control systems:
1. The main controller LCD touch screen, can do programmable test
2. PLC complex man-machine interface control, contrived moving time, rest time and power-on time setting.
3. multiple sets of test procedures can be planned in the interface, and each group can be arbitrarily set the desired action, easy to operate.
4. If a fault occurs in the operation of various signals are written records will be displayed to facilitate maintenance.
5. Touch the setting, digital direct display, all the conditions fully automatic locking by a microcomputer.
6. imported electronic temperature and humidity sensor, not need to cleanup.
1. LCD Touch Screen Programmable Dust Test Machine Inner material: Stainless steel (SUS #304)
2. LCD Touch Screen Programmable Dust Test Machine Outer material: Powder painting steel plate
3. Dust Test Chamber Use imported touch screen programmable controller, can control dust blowing motor, dust vibration and test time.
4. With sealing good chamber door, has lock device and glass observation window.There is lamp and dust brush to ensure easy observation of inner chamber.
5. With strong dust blowing device and vibrator to ensure informity of each blowing.
6. Circulating air supplying device: Frequency converter and high pressure motor to ensure the air volume more precision.
7. Large diameter strong stainless steel air hose, covered with heating device, can heat and dry the test chamber dust.
8. Anti-explosion handle, easy operation, strurdy and durable.
9. Imported high-precision electronic temperature humidity sensor and air speed sensor.
10. Test sample differential pressure test device, can set pumping rate freely according to sample inquiry.
11. There is a ψ50mm test hole on the left chamber and equipped with port lid and silica gel chock plug. Through this hole can connect to outer test device.
12. There are casters with brake equipments and fixing foot pad to make the machine easy operation and level adjustment.
13. With voltage current indicator, equipped with corresponding sample DC12V power, DC24V / 20 a, interval, continuous power supply adjustable. Dust concentration wind speed regulating function. May control test total time, blow the dust of time interval and blowing dust cycle and sample with the power source of the intermittent time.
14. Vacuum system: Equipped with pressure meter, air filtration, dust dehumidification system, regulating triple pieces, connecting pipe (Vacuum pump should be prepare by customer)
|Contact Person :||Vickey Jin|
|Tel :||+86 13925868409|